Payment & Shipping Terms:
|Name:||Insulation Resistance Tester||Model:||HY2305|
Digital Megger Hand Driven Insulation Tester Multifunction Electrical Insulation Test Equipment
|(1)||EARTH||Connected to the shell of DUT or on the ground.|
|(2)||GUARD||Connected to HV retaining ring of DUT in order to eliminate the effection of surface leakage current..|
|(3)||LINE||High-voltage output terminal, connected to high-voltage conductor of DUT.|
|(4)||LCD Display||Display test results and setting parameters|
MEM (Memory):Saving test data or view historical test
data, keep pressing more than 3 seconds to delete all
the stored datas.
Time Setting: Setting Time using the button
Auto off (Backlite):Press the button to light up LCD
backlight, and automatically turns off after 90
Up:Press to forward read data or modify data;
Down:Press to back read data or modify data
|(6)||INDICATOR||Showing HV output, power supply working,charging ...|
|(7)||TEST KEY||Press Start testing, clockwise rotation can be locked after pressing the button|
|(8)||CHARGING JACK||15-19V DC|
|(9)||SHIFT||Selecting Output voltage, also power switch|
Method of Usage
Note: When you use the instrument for the first time, it is asked to need charging 6 hours otherwise the instrument may does not work normally. Charging method see chapter of Battery in the below relevant content.
1) Please cut power supply and remove any outgoing connctions to DUT (Device under test) before testing, and DUT will be shorted to ground for discharging 1min, DUT of larger capacity should be discharging at least 2min in order to avoid electric shock and impact of the discharge measurements.
2) With a dry soft cloth to wipe clean the dirt surface, if necessary, wash with gasoline before fouling the surface of the casing in order to eliminate surface leakage current of test results.
3) High Voltage testing wires (red) put into position (3)-LINE, hook the other end connected high voltage conductor, testing line (green) put into position (2) -GUARD, the other end to be connected to the HV retaining ring of DUT to eliminate the impact of surface leakage current (see "GUARD" description in related content), the other black test lead put into the earth terminal (1) - EARTH, the other end is connected to the shell of DUT or to the ground.
Note: In order to avoid overload, the wiring of GUARD is strictly prohibited to connecting with LINE (short-circuit).
Contact Person: Chloe